|Introduction / Operating Information|
The steps to perform to ensure the equipment is working correctly at the start of a shift.
This section of the Directions For Use (DFU) is intended to describe the checkout steps required for safety and effectiveness in using this medical device.
At the start of each shift, staff should check to be sure the equipment is working correctly before they begin to perform their tasks on the unit. The nursing staff should verify that their assigned mobile device is working correctly. Staff with management responsibility should ensure that the web browser is functioning as required. Contact support for assistance if problems occur.
Checkout procedures for the user consist of the following steps.
Send a test alert and check the time between alarm initiation on the medical device equipment and the alert receipt on the mobile device. Total latency should be under 5 seconds.